Find the most up-to-date version of DIN at Engineering DIN Field procedure for precision testing of surveying instruments; general information. DIN Field procedure for precision testing of surveying instruments; gyroscopes. standard by Deutsches Institut Fur Normung E.V.
|Published (Last):||6 June 2007|
|PDF File Size:||18.63 Mb|
|ePub File Size:||17.31 Mb|
|Price:||Free* [*Free Regsitration Required]|
The error in an angle is given by the equation: The first one is for the “testing theodolites” and the second is for levels.
You couldn’t apply the findings of that one gun and apply it to the rest of the guns of the same make and model in your company.
August 3, 7: All these 2″ were taken during lab conditions where everything is controlled.
I was born working and I worked my way up by hard work. Space Weather influence on satellite based navigation and precise positioning R.
DIN and in English? : Surveying
On hillsides- one leg uphill and two legs downhill If the diin in soft or muddy, drive hubs in the ground to support the tripod legs. The optical plummet can be checked by rotating the knurled eyepiece until the cross hairs are the thickest and blackest, refocus and check for parallax.
Kent, I guess you forget that you are an anomaly. Usually set up within 0. I mean really know all the math dim it?
See below for screen shot of my data sheet of my instrument. The standard used to determine accuracy compared to the true value.
I would think this is a more complex version of taking an instrument to a baseline and figuring rin its tolerances.
My presentations Profile Feedback Log out. Paul in PA, post: August 3, 8: What is the estimated error in the angle due to pointing and reading?
The document is titled: Forum contains no unread posts Forum contains unread posts Mark all read. I have the 12th edition at hand. That’s the article I would have pointed you at, and realistically all you need to know, unless you’re doing some sort of higher order stuff.
If you dun at the title of DIN its: This type of error will appear as a random in the adjustment of a network involving many stations. Without paying 60eur to read it, I’m not going to go any further. Submit a new link. Do you have any information to back that up? Or, in the case of Leica, using a circle with known periodic errors in its graduations that is either partially 81723 entirely removed by the reading system vin, depending upon the instrument accuracy spec. So, for a 2″ Din spec instrument: August 3, DIN loosely translated German Institute for standards.
In other words the difference in price buys the exact same instrument, but with a few more lines of code active in the internal software.
Land Surveying Community Forums
True Location Dependent on quality of instrument State of adjustment of optical plummet Skill of observer Can be compensating Error is maximized when the individual setup is on the angle bisector. Least count- with the advent of electronic instruments, reliance on the least count for anything but an estimate of precision achievable not accuracy is highly inadvisable.
A Worldwide Repair Service – Visit us at: